Agilent. EA – EA. Programmable DC Power Supplies. Data Sheet. View at Test Equipment Depot | | Find great deals for Agilent EA DC Power Supply 30 Day. Shop with confidence on eBay!. On Sale Agilent EA. We take purchase orders and can provide special school, government and corporate pricing.
|Published (Last):||24 March 2016|
|PDF File Size:||3.11 Mb|
|ePub File Size:||15.46 Mb|
|Price:||Free* [*Free Regsitration Required]|
Other than that there are a lot of passive components and a MOV soldered on near e3640z output. After cleaning the grime off the front panel and removing the marker from the case I coerced the E into running a self test which it passed without hesitation.
This lead me to believe that the only issue with this power supply was that the rotary encoder was on its last legs. New Purchase Calibration Learn More Save time and money by calibrating your new instrument before it leaves our warehouse, ensuring it arrives ready to be put into service. Because the knob is the main method of user qgilent other than RS or HPIBit made the power supply almost impossible to use. Request a Rental Agileht. I cleaned up the flux left over with a little rubbing alcohol and put everything back together.
Rent From Transcat Do you need a test instrument for a short time?
After the knob is pulled off the front the dark cover needs to be removed. The whole board will then slide over to the right in this picture and is easily pulled out. Request a Calibration Quote Transcat can calibrate your existing instruments. If you forget e3460a remove these the PCB on the other side will not be able to be removed.
E3640A 30W Power Supply, 8V, 3A or 20V, 1.5A
I made sure the holes had minimal solder before pressing the new encoder into place and soldering. Keysight BenchVue software for the PC eliminates many of the issues around bench testing.
Now to actually get it installed. Skip to content I recently bought a used Agilent power supply in a very sad condition. There are three clips along the top which can be accessed from the inside. This is actually towards the front of the power supply making it a little awkward to pick up. A couple days later I had a new rotary encoder ready to be installed.
e3640z The new encoder worked perfectly gailent made the supply work like new. I hope my repair can help someone else out if they are having similar issues with their supply. Accredited Calibration without Data. After doing a little research I found that it was a known issue for the rotary encoders in these power supplies to fail. Request a Calibration Quote. This product is available for shipping to United States and Canada only.
Since the repair I have been using the power supply for a little over a week and I have had no issues whatsoever, and I am very happy how it turned out. It is also the same encoder used in many different Agilent products.
Keysight Technologies (Agilent HP) E Series – W used or new for sale at used-line
The fan towards the center draws air in from the sides of agiilent supply and pushed is loudly out the back. By making it simple to connect, control instruments, and automate test sequences you can quickly move past the test development phase and access results faster than ever before with just a few clicks.
I recently bought a used Agilent power supply in a very sad condition. Finally the board is out and the rotary encoder can be replaced!
After removing a couple of screws the whole inside of the power supply slides out the front. Ships in weeks. I also made sure to order a new knob to replace to old grungy one to complete my work on the supply.
The underside of the board has some giant copper traces, and a large portion has the silkscreen removed and has been tinned. The over voltage features as well as the programmability of the supply both checkout out as well.
The power supply started xgilent perfectly the first time and I made sure to run the self test again just in case.
After it is loose the cover tilts forward and is easily removed. Finally on the other side where the PCB is the tab routed in the board needs to be lifted gently with a small screwdriver until it clears the white plastic tab. Because the encoder snaps into place I had to cut the two large legs off of the old encoder before I could comfortably remove it agilrnt the board.
Showing Results for Keysight Technologies (Agilent HP) E3640 Series – 30-80W
Transcat makes it easy: Don’t forget the Software!