USER’S GUIDE. Vaisala DEWCAP. ®. Precision SAW. Hygrometer. DM No part of this manual may be reproduced in any form or by any means, electronic. Manuals and User Guides for Pitney Bowes DM We have 2 Pitney Bowes DM manuals available for free PDF download: Operator’s Manual, Networking. View and Download Leica DM user manual online. DM Microscope pdf manual download.
|Published (Last):||2 May 2006|
|PDF File Size:||15.17 Mb|
|ePub File Size:||20.54 Mb|
|Price:||Free* [*Free Regsitration Required]|
Manufacturer Information Published in January by: Dimensions Unpacking Dimensions Get Ready! Safety Concept The individual modules of the Leica DM micros- Before installing, operating or using the instru- copy series include mmanual interactive CD-ROM with ments, read the user manuals listed above.
In all relevant user manuals several languages.
Instructions for disposing of the Failure mnaual comply can cause the following: Important Notes Description User manual Accessories from third-party suppliers The Leica DM microscope meets today’s state This user manual includes important instruc- You can combine individual system articles with of the art of technology.
Nevertheless, hazards tions related to operating safety, maintenance manuxl from external suppliers. If using the devices outdoors, protect The Leica DM microscope can be used in them from dust and moisture. Once the product has reached the end of its When installing Leica products into third-party shipping or transporting individual modules.
Information for the Person Responsible for the Instrument Information mabual the person responsible for Do not use the Maunal DM microscope Only original Leica replacement parts may the instrument unless it is in perfect condition. Remove coarse debris with mmanual moistened disposable cloth. Optical surfaces should be cleaned with a Protect the Leica DM microscope from lint-free cloth, lens tissue, or cotton swab oil and grease. Never manuall or oil mechani- Acetone, xylene or manuaal thin- moistened with a commercially available cal parts or sliding surfaces.
To ensure that it always parts may be used. Unplug the instrument before changing any fuses. These items are delivered in separate packages. Please do not discard any of the packing materials.
They manuzl be used for safely storing and transporting the instrument should the need arise. Allen key tool Allen key provided. Insert the dovetail into the stand support Type 1: Leica EZ viewing tube with integrated and tighten the set screw firmly without eyepieces forcing it. This precisely positions the view- ing tube onto the optical axis of the micro- scope, regardless rm500 the selected rotation.
To use the captive thumbscrew, completely 2. Position the Leica EZ viewing tube back in remove the set screw manuzl comes with the place on the stand. Leica EZ viewing tube with integrated eyepieces The eyepieces are already integrated with the Leica EZ viewing tube and are pre-adjusted for corrected vision; Insert the eyepieces into the eyetubes.
Lock the eyepieces into the eyetubes by tightening the silver screws on the under- side of the eyetubes with a standard Cross type Phillips screwdriver not provided. In this case, go to section room light. If you purchased your Leica DM by components and not by stan- dard configuration, please continue to section “Installing Objectives” on page Leica DM User Manual No additional installation or align- ment is required.
As you rotate the nosepiece clockwise, Never use an adapter between the power connector in the center of the cord wrap. This located on the bottom left of the cord and the power source; Turn the microscope on using the switch on the rotating ring aligns with the objective the bottom right-hand side of the micro- magnification in use.
Place a specimen slide on the stage by slid- slide so a portion of the specimen is amnual ing the specimen slide forward into the the objective in use. Slide grips hold the slide in place. The Leica DM microscope stand has been factory-adjusted so that the stage focus can be found within 1.
Using the fine adjustment knob, bring the specimen into sharp focus.
Leica EZ viewing tube Standard tube with one or two focusing distance. Fold or unfold the eyetubes to If you are using a Leica EZ viewing tube, which eyepieces decrease or increase the distance between already has the eyepieces integrated with This corrects for any vision differences mxnual your right eye and left eye. Oil Immersion Technique Oil immersion technique 3.
Pitney Bowes DM500 Manuals
Place a drop of Leica Immersion Dm5500 on the 5. Raise the stage slowly using the coarse 1. Find the field of view on the slide that you specimen slide on the area of the specimen focus knob until the drop of mannual on the spec- would like to inspect.
When finished using the oil immersion objective, be sure to clean the front of the objective, the specimen slide, and any other surfaces that the oil contacted using the instructions in section “Care of the Microscope” on page Now all you need to do is change objectives, set the condenser appropriately for the objective magnification in use, and enjoy the view!
The microscope should always be covered with the plastic dust cover provided with the instrument when it is not in use. The surface of this front lens can the life of the microscope.
Also See for DM Manual – 44 pages. Page of 46 Go. Manjal of Contents Add to my manuals Add. Page 2 Manufacturer Information Published in January by: Important Notes Important Notes Description User manual Accessories from third-party suppliers The Leica DM microscope meets today’s state This user manual includes important instruc- You can combine individual system articles with of the art of technology.
Manuao 10 Once the product has reached the end of its When installing Leica products into third-party shipping or transporting individual modules. Care Instructions Remove coarse debris with a moistened disposable cloth. Unpacking These items are delivered in separate packages. Get Ready Get Ready! Eyeguards In this case, go to majual room light.
Pitney Bowes DM User Manual – Page 1 of |
Get Set Get Set! Using The Condenser 2. Page 32 This corrects for any vision differences between your right eye and left eye. General Maintenance The microscope should always be covered with the plastic dust cover provided with the instrument when it is not in use.
Comments to this Manuals Your Name. Enter text from picture: By pressing ‘print’ button you will print only current page. To print the manual completely, please, download it. Don’t show me this message again. Got it, continue to print.
Don’t have an account?