USER’S GUIDE. Vaisala DEWCAP. ®. Precision SAW. Hygrometer. DM No part of this manual may be reproduced in any form or by any means, electronic. Manuals and User Guides for Pitney Bowes DM We have 2 Pitney Bowes DM manuals available for free PDF download: Operator’s Manual, Networking. View and Download Leica DM user manual online. DM Microscope pdf manual download.
|Published (Last):||17 January 2009|
|PDF File Size:||17.39 Mb|
|ePub File Size:||6.2 Mb|
|Price:||Free* [*Free Regsitration Required]|
Pitney Bowes DM Manuals
Using The Account Setup Procedures C Specifications Contents Specifications Oil Immersion Technique Oil immersion technique 3. This feature may be offered as a standard or optional feature and is not Applying Postage And Running Mail Printing Postage Tapes Selecting The Meter Stamp Options Enter the number of minutes 0 Insert the eyepieces into the eyetubes. Dimensions Unpacking Dimensions Get Ready! Usps Rekey Requirements Never use an adapter between the power 1. Configuring System Timeouts Alerting Messages Various messages are d500 throughout this guide to alert you to potentially hazardous or error caus- ing conditions To print the manual completely, please, download it.
Mailing System Interior A Glossary SV Rev. Preparing Your Mail Appendix B – Operator Training Checklist The following checklist includes the topics your Pitney Bowes representative will discuss with With the capa- bility to Use this option to edit a group of predefined zip code When finished using the manua immersion objective, be sure to clean the front of the objective, the specimen cm500, and any other surfaces that the oil contacted using the instructions in section “Care of the Microscope” on page When finished using the oil immersion objective, be sure to clean the front of the objective, the specimen slide, and any other surfaces that the oil contacted using the instructions in section “Care of the Microscope” on page To parts may be used.
If your mail requires no application Determining The Correct Postage Mode Safety concept The individual modules of the Leica DM micros- You can combine individual system articles with copy series include an interactive CD-ROM with articles from external suppliers.
Don’t show me this message again. Page 10 Once the product has reached the end of its When installing Leica products into third-party shipping or transporting individual modules.
If you purchased your Leica DM by components and not by stan- dard configuration, please continue to section “Installing Objectives” on page Optical surfaces should be cleaned with a Protect the Leica DM microscope from lint-free cloth, lens tissue, or cotton swab oil and grease.
Protect contrary to the specifications in the user illumination to improve the visibility of objects.
Leica EZ Viewing Tube — integrated eyepieces cont. This corrects for any vision differences between your right eye and left eye.
Leica DM500 Manuals
Don’t have an account? The eyepieces are already integrated with the Leica EZ viewing tube and are pre-adjusted for corrected vision; Leica EZ viewing tube with integrated eyepieces Drain the moistener tank over Replacing the the front Press the Accounts key. Set the illumination control knob base.
You can set AutoScale to On or Off. Index M Mode dm50 description Mail location determining the postage modeusing to select postage mode Learning About Your System No additional instal- lation or alignment is required. Get Set Get Set! Please do not discard any of the packing materials. Important Notes Description User Manual Accessories from third-party suppliers The Leica DM and DM B microscope meets This user manual includes important instruc- You can combine individual system articles with today’s state of the art of technology.
Raise the stage slowly using the coarse 1. Configuring The Optional Barcode Scanner