Read online or download PDF • Page 44 / 67 • KROHNE IFC Converter EN User Manual • KROHNE Equipment. Micro Precision provides expert calibration services for KROHNE instruments including KROHNE IFC D -. Select from options for the IFC D which. KROHNE 08/ Subject to change without notice. Applicable to. Software-Versions. ○ IFC _. / D. Display version. No. and.
|Published (Last):||15 December 2008|
|PDF File Size:||9.84 Mb|
|ePub File Size:||5.2 Mb|
|Price:||Free* [*Free Regsitration Required]|
The operator is, howeverobligated to take note of these sections in the. The selected creepage distances and clearances have been sized in conformity with VDE and IEC for contamination category 2. Page 15 Function 3. Signal cable to be permanently laid.
Setting ranges The ranges are dependent on the meter size DN and dd flow velocity v: Tell us about it. Connection of shields to the IFC 1 DS Signal cable 2 Shield terminal for shielded signal cable 3 Field power cable 4 Shield terminal for shielded field power cable 5 Housing wall, signal converter 6 Cable entries 2 Electrical connection of outputs 2.
100 el measuring instruments. All operating data have been factory set in accordance with your specifications. Tell us what’s missing.
Switches, counters, displa ys and recorders. Your manual failed to upload Vo r tex flowmeters.
KROHNE Aquaflux K/D/6 with Signalkonverter IFC D Unused | eBay
Do not cross or loop cables inside the terminal compartment. Use separate cable entries see below for power supply, field current cables, signal lines, outputs and inputs. Page 14 Function 1. Don’t see a manual you are looking for?
KROHNE IFC 010 Converter Short EN User Manual
Or point us to the URL where the manual is located. Page 10 3 Start-up 3. Supply circuits and output circuits are designed to meet the standards of overvoltage classes III and II, respectively. V ariable area flowmeters.
If installed and operated in accordance with these operating instructions, it will rarely present any problems. Page 2 Contents 1 1.
KROHNE IFC Converter EN User Manual | Page 44 / 67
P Pulse output overranged. Page 8 Connection of shields to the IFC 0110 DS Signal cable 2 Shield terminal for shielded signal cable 3 Field power cable 4 Shield terminal for shielded field power cable 5 Housing wall, signal converter 6 Cable entries 2 Electrical connection of outputs 2. Passive mode External power source required for operation of the outputs.
The flowmeter housings protecting the electronic equipment from dust and moisture must always be kept closed. Power the unit, and the flowmeter will immediately start process flow measurement. Connect shields via stranded drain wires.
Underwater and underground installation possible. Page 11 Standard factory settings Function 1. These concise instructions do not include the following: Page 3 1 Electrical connection: